Posted in the Agilent Technologies Forum
#1 Mar 4, 2008
I have a problem with my Network Analyzer so where can go with my Problem.
The measurement of My Missmatches (Amritsu 20 dB and 6 dB) are out of Range above 3 GHz!
I have made a verfication and a calibration!
By the PNA Operator Check
Prompt for attachment of short/open
that is getting a failed check!
So where can i ask for support?
thanks in advance!
Add your comments below
|Agilent Honors RS Components (Feb '11)||Feb '11||Ng Wan Hoe||1|
|Startup incubator fosters growth (May '10)||May '10||North America||2|
|Agilent dips $40M in new India campus (Apr '10)||Apr '10||Agilent India emp...||1|
|error 538 Agilent 8561E (Dec '09)||Dec '09||domlaco||1|
|Napolitano outlines border security plans (Apr '09)||Apr '09||Ron Adams||5|
|Agilent laying off 2,700, halting stock buyback... (Mar '09)||Mar '09||Ben Dover||1|
|Agilent High Speed Lightwave Component Analysis... (Jul '08)||Jul '08||boliang||1|
Find what you want!
Search Agilent Technologies Forum Now
Copyright © 2015 Topix LLC